Hi I have a thermo scientific Epure 4 module water filtration system model D4641 7119 The resistivity meter was reading 182 on a saturday morning while it was being used The filter system was left on over the weekend and monday morning the resistivity meter was reading 24 My question is how do I troubleshoot this problem Is is the meter or the cartridges Will cartridges go from 182 to 24 that quickly Thank you
Turn the system on and draw off water and see if the resistivity goes up. If it climbs way up when drawing water, but is low when not drawing water, usually means you have a bad pump. If it stays low or goes lower when drawing off water, the filters need to be changed. It is normal for the system to crash quickly when it does.
Great for Lab Water Pretreatment
or Great Drinking Water
Images are representative of the products. Images may or may not be of the actual product. If it is important e-mail us for an actual image if available.
* Flat Rate UPS shipping when able to ship via UPS and is in the USA excluding Hawaii and Alaska. Larger Items may not be able to ship via UPS, in that case freight charges will be quoted seperately. International shipping will be quoted after the order is placed. You will have the opportunity to cancel before we finalize your order.